Q. How many test sessions can be stored on the GSI TympStar Pro?
A. The last 100 test sessions are automatically saved to the TympStar Pro.
Q. How do I use the Auto Start with the shoulder box?
A. For both screening and diagnostic mode, insert the probe and as soon as the seal is obtained, the sweep begins without any need to press additional buttons on the instrument or probe.
Q. What probe tone testing frequencies should I use with the TympStar Pro?
A. The standard multiple probe tone frequencies are included: 226, 678, and 1000 Hz. The 1000 Hz probe tone frequency allows the audiologist to accurately conduct tympanometry testing on infants up to 6 months of age. 678 Hz helps confirm the diagnosis of ossicular discontinuity, otosclerosis, and other ossicular chain disorders. 226 Hz is the probe tone used most commonly. It is used to perform tympanometry, reflexes, reflex decay, and other diagnostic tests on patients 6 months and older.
Q. What type of programmable user tests are available with the TympStar Pro?
A. Audiologists have the ability to set the deflection criterion for reflex threshold testing, pump speed, sweep direction, stimulus starting intensity, and more.
Q. What is the EMR/EHR compatibility with this tympanometer?
A. The TympStar Pro can offer e-record solutions for all types of facilities. Reports may be automatically stored in compatible format to attach to EMR or a true HL7 interface with AudBase, Greenway, or AllMeds EMR.